For Windows NT, change references to win95 to winnt. But I get an error message actually on the fg when the initialise vi is run in Labview. If you do find a command that is causing the instrument hp a labview report an error, contact NI so that they can fix the bug. Mean time between failure MTBF is 40, hours. If the NI driver does not handle these two cases differently, I can guarentee that you will have problems.
|Date Added:||24 January 2018|
|File Size:||55.89 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
It will be tough for me to adjust this one.
HP A driver problems – Discussion Forums – National Instruments
It should be able to tell you what commands are really being sent. Message 6 of All functions of the hpA. While looking up other things, I found: To avoid this, I decreased the output range of the function generator so as not to activate the switch.
Changing output amplitude on Agilent function generator while output is on Solved! Hp a labview help file presents application programming examples, a cross-reference between instrument commands and driver functions, and detailed documentation of each function, with examples.
To load the dll you must include the file hpa.
I don’t remember about theso I can’t guarentee that the long form works. Comparing the tick count to you wait seems dangerous to me.
Keysight Technologies / Agilent Technologies hp33120a Signal Generator
What is inside of the VIs that set the amplitude? Mean time to repair MTTR is 1. Thanks for the suggestion Mateus, but I am trying to automate this process as best I can.
It includes a “Function Panel”. Source is included so that the driver can be modified if desired. If someone has access to the dll source code, could you please list the init function?
So I know the abbreviated version works.
The help file presents application programming examples, a cross-reference between instrument commands and driver functions, and detailed documentation of each function, with examples. Now the waveform changes amplitude without turning off the output.
In this example, serial communication is used. I opened the Configure Standard Waveform block and this is what it looks like: Introduction The help file included with the HP A driver contains instrument labviea programming information. If you do find a command that is causing the instrument to report an error, contact NI so that they can fix the bug. Is the code labvieq hpA.
Thanks for the help answering my questions.
HPA Improper waveform output when controll | Keysight Community
Message 2 of Implementation of the addressing scheme is vendor specific and some vendors support mixed cases. Transportability across different computer platforms 2. Other requirements for successful RS communication with the It includes a Visual Basic include file. Remove the fuse-holder assembly with the blown fuse from the rear, and install the new fuse, Then the Configure Hp a labview VI is called to set the instrument up for the desired measurement.
REM” after the reset. It is likely they turn off the output and turn it back on.
I opened the Configure Standard Waveform block and this is what it looks like:.